Improved design of the adapter for more stability of the endoscope and smartphone. Now the arms supporting the endoscope are longer with rubber pads. This allows more security and stability in holding the endoscope. Curved arms supporting the smartphone.
Now its almost impossible to slip from the smartphone. Improved quality of the material of the adapter. Portable LED light source features double the light intensity than most portable light sources in the market.Now it can be used safely in long surgeries without being worried about battery drainage, thanks to the available external USB power supply. Micro USB power and charging slot makes it easy to charge any time any where with any USB chrging slot. The micro USB slot is not only used for charging, but also for powering the light source, so you do not have to worry about battery drainage in long surgeries.
Unlike the mechanical button switch that ruins with time in most portable light sources in the market, the. Smart portable LED light source features a Smart switch button and controller with LED indicator which allows for light intensity control and also gives data about battery charge level and charging status.
Standard USB slot for using this light source as a power bank for emergency smartphone charging. The included battery is a genuine Panasonic. Battery with a trusted capacity that lasts more than 3 hours "according to light intensity used" and long life span with more than 500 cycles of usage. Dimming circuit for optimal operating light intensity control. Smart circuit that stops charging when charging complete and stops the device when battery is almost empty, so attaining maximum battery protection with longer life span.
An accessory adapter for connecting Olympus endoscopes and also light cables with Olympus connection. Connect the smartphone adapter with the smartphone and the endoscope. We recommend "open camera" app for android phones and "eipro" app for iphone models. These apps allow manual focus control and more advanced camera control features.A small circle of the endoscope image will appear on the screen and you will have to zoom in to fill the smartphone screen with the endoscope image. To expose the USB charging slot, rotate the head of the light source counter clockwise. The LED indicator of the button switch turns red when charging is in progress and turns blue when charging is complete. Now you can disconnect the charger and start using the device. Connect the endoscope or light cable to the light source. The light will turn on with moderate intensity (this is the recommended intensity for most of examination and long surgeries). Press the power button again within 2 seconds>. The light will turn to maximum intensity (this mode is ideal for short procedures that requires intense light - not to be used for more than 10 minutes).
Pressing the power button after 2 seconds from turning on will turn off the device. When the device is on in any of the above modes, long press the power button with start the dimming feature. Leave the button to keep the desired light intensity. Rapid 2 press will turn strobe feature on (beta). The LED indicator in the power button turns blue when battery is charged, red when it gets weak and flashes red when the battery is empty before the device turns off.For endoscopes with Storz adapter, please remove the adapter first, then connect the light source to the inner thread of the endoscope. 1- The maximum light intensity mode is only suitable for short procedures, and it should not be turned on for more that 10 minutes to avoid over heating of the device. For longer procedures, please turn into the medium light intensity mode. 3- The quality of the produced image depends mainly on two factors: The diameter of the endoscope and the smartphone camera quality. 5- Please do not use the built in android camera app "in old android phones" because it doesn't provide white balance, exposure or focus control which might be deceiving that your smartphone is not compatible, but "open camera" app gives best results. Please, fully charge the battery before using it.
8- For longer battery life, please do not full charge or fully discharge the battery. 9- If the device is to be stored for a long time, please keep the battery in a partially charged state (about 50%) and do not store it when it is discharged. USB Output power: 5V 1A. Emitter BIN: T6 up to 10 W.
Color BIN: neutral White (daylight). Working Voltage: 3--4.2 V. Power Supply: 1 x 18650. Actual Lumens: up to 1000 lumens. Run time: 3-5 Hours (depending on light intensity used).Number of Modes: one press to boot at moderate intensity - press again within 2 seconds gives maximum light - long press 2 seconds: 10%-100%. Quickly press the two times to enter strobe mode(beta). Switch Type: Smart touch switch with LED indicator. Dimensions: 13.5 cm x 2.6 cm. Weight: 134 g with battery inside.
Portable LED light source (endoscope not included). 3-Accessory adapter for connecting Olympus endoscope and Olympus light cables. International Buyers - Please Note. This item is in the category "Business & Industrial\Healthcare, Lab & Dental\Handpieces & Instruments\Endoscopes". The seller is "claramed" and is located in this country: EG.This item can be shipped worldwide.